to set up the software for your use. This guide shows a workflow that uses application settings. BD FACSDiva Software Quick Reference Guide for the BD LSR II. Compensation Controls in the BD FACSDiva Software Reference Manual. • Default templates are now provided for certain instrument functions. If you are the first user of the day: Boot up the computer and log on to Windows. Wait 10 seconds, then Open the BD FACSDiva software and log on. Allow the.
|Published (Last):||27 September 2009|
|PDF File Size:||3.58 Mb|
|ePub File Size:||9.46 Mb|
|Price:||Free* [*Free Regsitration Required]|
Messages are listed next to the time the event occurred.
Press Enter to apply the new name. Updated features are highlighted in the New Features section. The computer is left on at all times.
Getting Started with BD FACSDiva Software
Summary of the Upgrade Process on page 6. You must first assign a detector before you can enter a parameter name.
CompanyName 3 Click Save to save your changes and dismiss the dialog. You referencf close an open experiment during acquisition. These fields cannot be changed. The hierarchy in the Browser has to be followed when moving elements: InfoPrint Serial Matrix Printers. To make sure the duplicated analysis starts a new page, move or edit worksheet elements so they fit a full page before you duplicate the Analysis.
BD FACSDiva Software Reference Manual
The Statistics checkbox exports statistics to a single file for the batch; when the Statistics checkbox is deselected, the Export Filename field and its Browse button are unavailable. Folders can be used to group experiments. No part of this guide may be reproduced or distributed, More information. See Changing Plot Parameters on page Instrument and Acquisition Controls When a parameter exists in the target, but not in the source, the values for that parameter will not be changed.
Select a category from the menu; then enter specific information in the text field next to the menu. Gates in the software-defined analysis objects are snap-to gates, so minimal editing is required. Figure Example organization facsdva daily work ; Tip List experiments by date by clicking the Date column header in the Browser, or list experiments alphabetically by clicking in the Name column header.
See Using the Tube Inspector on page Tip Extend the gate past the plot axes to capture events on the axis. Erference Working with the Biexponential Editor on page See Disabling Users on page Senior Systems Cloud Services In this guide We cannot be held liable More information.
In this case, the new population is a subset of P2 because it is highlighted, even though the plot is showing All Events.
Page of 6 Go. Each selected element will be stored as a separate JPEG file.
BD FACSDiva Software Reference Manual |
The Experiment Templates dialog box appears where you can select the template type and number of experiments to create Figure For example, sorting the Q1 cells would result in the following expression being sent to the cytometer: Before Using This Machine.
The P5 gate label is marked with two vertical bars, indicating that it is the population for which scaling is in effect. Use these buttons to align worksheet elements at their left, right, top, and bottom edges, respectively, with the last-selected object object with yellow selection handles.
Note that when you assign an analysis template as the default global worksheet, the assigned template is added to each new blank experiment. Microsoft and Windows are registered trademarks of Microsoft Corporation. Refer to your instrument manual for details.